Blind study compares hearing aid and PSAP sound quality and speech intelligibility.
In recent years, a new category of hearing amplification devices has been introduced. Although not intended to compensate for impaired hearing,Personal Sound Amplification Products (PSAPs) have been proposed as a cheaper alternative to hearing aids, and some studies have concluded that PSAPs, in select audiological domains, offer the same benefits as conventional hearing aids for people with mild-to-moderate hearing losses. However, very few of the published studies have compared the two product categories while ensuring proper test-subject blinding, and (to our knowledge) none have ensured test-subject blinding and individualized fit of the hearing devices.
Most PSAPs are being marketed and sold as over-the-counter (OTC) devices, typically purchased online. Some devices are fitted by the end-user at home, and some devices are not fitted at all. Thus, a major factor contributing to the lower retail price of PSAPs is that the end-user does not pay for the service of a hearing care professional. One disadvantage of this is an often-imprecise fit of the device to the individual hearing loss.Despite this and the “basic level” technology of most PSAPs, it has been argued at conferences that, for people with mild-to-moderate hearing losses, some PSAPs can offer an audiological performance similar to traditional hearing aids. However, when one considers the amount of time usually spent on fitting and fine-tuning conventional hearing aids, as well as the amount of resources required for the development of advanced hearing aids, these findings are quite surprising.
This study investigates whether it is really true that there are no differences in the audiological performance of PSAPs and hearing aids when both are fitted in as standardized a manner possible, and when the experimenters ensure that the participants are properly blinded relative to the device being listened to.
Amplification devices. This study evaluated two premium PSAPs: Perfect Choice HD (PC) and Sound World Solutions CS50+ (SWS) against the Oticon Alta2 Pro (Alta2) hearing aid. Both are among the premium PSAPs, priced in the high-end of the market; SWS got good reviews by Mamo et al in a 2016 study and were reported to produce good listening performance by Reed et al in a 2015 study. Thus, these two products were chosen as representatives of some of the best PSAPs in an admittedly wide field of marketed products.
Test set-up. The test set-up. Experiments took place inside and outside an anechoic room. An artificial head (Klangfinder HS8 Pro) with four pairs of “ear canals” was placed inside the anechoic room. The artificial head was wearing a pair of all three devices (Figure shown). The participant sat outside the anechoic room and listened through headphones. In this way, all three sets of devices were operating at all times, and a simple switch could be made between the three pairs of devices while the participant was listening.
This setup ensured that the participants were blinded to the device they were listening through, thus avoiding potential placebo effects. Further, it allowed for a smooth and quick way to switch between devices, which is particularly important when comparing sound quality.
Fitting of the hearing devices. An important decision when designing an experiment aimed to compare the performance of different hearing devices is how they should be fitted to the individual participant. This study chose to make the fitting of all hearing devices as consistent and simple as possible. This meant that the Alta2 hearing aid was fitted in prescribed mode (VAC+, one program) with no fine-tuning and no REM verification allowed. This choice was designed to mimic the most basic (ie, least possible) fit in a real-life hearing care practice—and to avoid being accused of creating a hearing-aid biased study.
For the PSAPs, the same fitting strategy meant that they were fitted based on the best possible tool available to an end-user ordering the devices online. For the SWS, the programming was based on a “prescribed” setting obtained by running the custom-made hearing test available in the accompanying app, and by selecting the most generally used program: Program 1 (“Everyday”). The PC does not come with a hearing test, and selection of the program number (Programs 1 to 3) is the only way to influence the frequency response. Again, the purported generally used program, Program 1 (“Speech”), was selected.
In two separate experiments—called Experiments A and B—10 (A) and 11 (B) participants with mild-to-moderate hearing loss were tested using the Danish Hearing-In-Noise Test (HINT). This was done in four different spatial setups. The target speaker was always placed in front (0°), whereas the positions and types of the maskers varied.
Sound quality was also assessed in both Experiment A and B; however, the sound samples were changed between the two visits. In Experiment A, the subjects listened to three different pieces of music (classical, rock, jazz) and two types of speech (speech in quiet, dialogue in a cafeteria), whereas the subjects in Experiment B listened to two types of music (classical, jazz), three types of speech (speech in quiet, dialogue in traffic, dialogue in cafeteria), and finally to a soft signal of the chirping of birds in a forest. The classical and jazz sound samples were not the same in the two experiments.
Overall, these findings showed that the hearing aid produced much better speech intelligibility in all the non-standard spatial configurations. Even in the standard HINT, a significant performance difference was observed between the hearing aid and PC and between the hearing aid and SWS in Experiments. Further, in the restaurant scene (±135°, four-talker), where adaptive directionality was applied for some of the subjects (ie, only for those performing below 0 dB SNR), an even larger difference in performance was observed between the hearing aid and the best-performing PSAP.
In this study, the performance of two PSAPs were evaluated against the performance of a hearing aid for listeners with mild-moderate hearing losses. On the most important domains, speech intelligibility, it was shown that the hearing aid performed significantly better than both PSAPs. Relative to sound quality, the hearing aid was also on average preferred over both PSAPs, although one PSAP (SWS) was preferred equally often as the hearing aid on a subset of sound samples. Thus, it seems fair to conclude that, on speech intelligibility and sound quality, the hearing aid is the better rehabilitative choice.
Source: Information in this article was originally presented in a poster session at the World Congress of Audiology held during September 2016 in Vancouver, Canada by Filip M. Rønne, MScEE, PhD, and Rikke Rossing, MA
Image credit: Filip M. Rønne, MScEE, PhD, and Rikke Rossing, MA
A Atlantic Hearing Aid Center opened its doors in January 1971 and has been serving the hearing community ever since! We have always been conveniently located in beautiful Fort Lauderdale, Florida at 2310 East Oakland Park Boulevard.
2310 E Oakland Park Blvd
Fort Lauderdale, Florida, 33306, United States
Copyright © 2015 AAtlanticHearingAidCenter.com. All Rights Reserved. Site Designed and Maintained By Huntpub.com